Teledyne LeCroy WaveAce 2012 Oscilloscope Driver
MHz bandwith oscilloscope Learn More MHz bandwidth oscilloscope Learn More. Add to Quote. WAVEACE · Teledyne Lecroy WAVEACE WAVEACE , Teledyne LeCroy, MHz Bench, Digital Oscilloscope USB Display 2 Channel CAT I V Record, Playback, Save, WAVEACE pdf. HDO AND HDO OSCILLOSCOPES. Combining Teledyne . The WaveAce combines long memory, any of ten popular oscilloscopes from Teledyne LeCroy,. Tektronix and Agilent .. Teledyne LeCroy, Inc. All rights reserved.
|Supported systems:||Windows XP, Windows Vista, Windows 7, Windows 7 64 bit, Windows 8, Windows 8 64 bit, Windows 10, Windows 10 64 bit|
|Price:||Free* [*Free Registration Required]|
Teledyne LeCroy WaveAce 2012 Oscilloscope Driver
LCRYa leading supplier of oscilloscopes and serial data test solutions, announced that it has expanded its popular WaveAce TM oscilloscope product line to include 4 channel models from 60 MHz to MHz and a new 2 channel, 40 MHz model. The WaveAce series is now available in 11 different models and all Teledyne LeCroy WaveAce 2012 Oscilloscope offer long memory, color displays, extensive measurement capabilities, and advanced triggering.
Combined with the streamlined, time-saving user interface, these features make the WaveAce oscilloscopes the ideal tools for affordable design, debug and troubleshooting from 40 Teledyne LeCroy WaveAce 2012 Oscilloscope to MHz. With the support of our solid distribution partnerships, LeCroy will now address a much larger segment of that market. With 32 built-in automated parameters, including advanced timing parameters for skew, phase and edge-to-edge measurements between channels, the WaveAce oscilloscope broadens the ways a user can understand and analyze waveforms.
About LeCroy Corporation LeCroy Corporation is a worldwide leader in serial data test solutions, creating advanced instruments that drive product innovation by quickly measuring, analyzing and verifying complex electronic signals.
The Company offers high-performance oscilloscopes, serial data analyzers and global communications protocol test solutions used by design engineers in the computer, semiconductor and consumer electronics, data storage, automotive and industrial, and military and aerospace markets. Quote Retarded memory: This was one of the WaveSurfer MXs models.
I probed a signal coming from an FPGA only to have it display an almost flat trace. It took me a couple of minutes to figure out that the sample memory was too shallow to allow any kind of decent measurement.
I then finally managed to figure out where to set a larger sample memory and just as I was thinking how stupid it was that the sample memory wasn't at its maximum setting by default, I realized why that was not the case. The scope became abysmally slow and only after a moment or two did the real waveform appear on the screen.
- Showing Results for LeCroy (Teledyne LeCroy) WaveAce 1012
- Teledyne LeCroy WaveAce 2012 100mhz 2gs/s 2ch. Oscilloscope 3 Year
- Teledyne LeCroy WaveAce 1012 (W1012) pictures
The WaveSurfer M Xs has automatic sample memory management which is enabled by default and which always uses enough memory to run the acquisition at the highest possible sample rate, so no, the low sample memory setting is certainly not the default for these scopes. It requires manual user action to change the memory settings, which suggests that Teledyne LeCroy WaveAce 2012 Oscilloscope you or someone else fiddled with the memory management settings, and I guess you probably also didn't bother to set the scope back to defaults before starting your measurements or someone has overwritten the default settings with their own.
That is user error, not a scope problem.
Teledyne LeCroy WaveAce 1000 and 2000 Oscilloscopes
And I can tell you that these things not only happen with LeCroy scopes. It also happens because, while EE's generally know very well how to operate a standard scope, many EE's simply fail to recognize that operating a modern high end scope and its advanced capabilities requires that some time is spent learning how to operate the scope properly, but many EE's simply fail to do that.
Your story above re. Again, why didn't you use Auto Setup? It would have corrected the Teledyne LeCroy WaveAce 2012 Oscilloscope immediately.
Quote It's not that they don't work well when you eventually manage to set them up right, it just takes a whole lot of time and frustration to set up something incredibly simple. It just kills productivity and irritates you without a good reason. I have Teledyne LeCroy WaveAce 2012 Oscilloscope the chance to play with many scopes over the years, but only LeCroys managed to piss me off every single time.
Teledyne LeCroy WaveAce / W Oscilloscopes Music instruments, Audio, Music
I'm sorry but so far I haven't heard anything that really justifies a statement like this: Quote from: Dave on February 08, What you've described looks very much about user error, i. That's not meant as offense, and I've seen many EE's struggling with operating advanced high end scopes especially when they normally use simpler scopes only.
But that applies to other scopes as well and is certainly not the fault of the scope.